


Technical Staff Links

FourStar Calibration
FourStar StandardsSpectrophotometric standards were convolved with the FourStar bandpasses to create a set of standard stars of appropriate brightness.
FourStar LinearityLinearity correction maps are provided for each array for the FullWell mode and the LoNoise mode. The filename is divided into 2 parts: the gain setting and the chip number. The deviation from linearity is a single term of order 2.5. The data were taken using dome flats with the H filter at varying exposure times scaled by interspersed 10s exposures. The data were fit as: exptime = A[0]*raw + A[1]*raw**2.5 + A[2] The coeffiecient A[0] represents the (inverse) linear gain of the pixel in units of [s/ADU]. The true expected counts are determined by dividing by A[0]: True counts = exptime/A[0] = raw + A[1]/A[0]*raw**2.5 + A[2]/A[0] The coeffiecient A[1]/A[0] represents the nonlinear correction and the term A[2]/A[0] is representative of the read noise (in ADU). These coefficients are stored in an image cube for each array. The cubes contain three layers: The first is A[0] which provides an indication of the pixeltopixel variation in linear gain, the second layer is the coefficient of nonlinearity of the 2.5th order and the third is zeropoint determined at the time of the fit. Since the zeropoint is presumambly random it is not included when applying the linearity correction. Since it is possible for raw CDS data to be negative in cases where the background is low, one can either use the absolute value when finding the half root or one can simply set a lower threshold above which to apply the linearity correction. In theory, the first frame (A[0]) can be normalized to unity and multiplied (since it represents inverse gain) with raw data frames to remove pixeltopixel variations in sensitivity. This would leave only flatfield variations in the telescopeplusfilter combination. Twilight flats contain all this information so it is not necessary to multiply by A[0], it is only mentioned here as a possibility.
FourStar Bad Pixel MasksThe bad pixel masks (bpm's) were created qualitatively using the linearity frames above and an IRAF cl script. The bps's for each chip are the same for either FullWell or LoNoise gain mode. The bpm's are shown here.


Carnegie Observatories, 813 Santa Barbara Street, Pasadena, California, 91101 USA The Carnegie Observatories  Las Campanas Observatories  Contact Us 